ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
|Published (Last):||24 July 2014|
|PDF File Size:||10.52 Mb|
|ePub File Size:||11.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
To activate the new settings: Page 48  that issues an alarm message at the display and an audible signal when the water falls below the minimal level. Icon description overview Figure. Chapter 4 Description of the device 4.
Hercaell the installation, check to see if the blower wheel is securely attached to the axle and if it can rotate freely, then secure cover using the screws. A pressure compensation opening with insert in the rear panel of the device allows a compensation between the pressures in the work space and in the operating room. Operational Safety Rules CO -incubator. Unless a user action contact with the pressure-sensitive areas and keys occurs within 30 seconds, the system automatically manusl the menu and the most recently confirmed setting defaults.
Page 38 For cleaning the interior, the recommendation is mild soap solution in water then rinse to remove residues. Install hose  to the sleeve of the gas humidification and then to the sleeve  of the device-integral oxygen or nitrogen supply line.
The comparison measurement must be performed when the device is completely heated up. Don’t have an account? Replacing the door seal Page 7 Magnetic door seal replacement Measuring cell with blower wheel and sensors 5. Page 66 The pipe of the pressure compensation opening must not be extended or redirected.
THERMO SCIENTIFIC HERACELL I OPERATING INSTRUCTIONS MANUAL Pdf Download.
If the device-integral monitoring system detects an error: O gas cylinder 7. Gas Connection Set the gas supply to a range between 0.
Then, position the lower rod guide  of the center strut into the hole  in the work space floor and remove spring tension. Chapter 10 Maintenance Press the power switch; the switch illumination goes 24i.
Thermo HERAcell-CO2-Incubator 150i 240i manual en
To exit the menu: Page Chapter 14 Device log Pressure compensation opening with insert 8. Operational Safety Rules The cross sections of the openings are smaller when samples are accessed. The work space of the device always remains visible, despite high humidity. I have 2 of 4 Heracell ‘s purchased in whose fan motors died within 1 month of each other.
Thermo Scientific Heracell i Manuals
No tools are required to replace the seal. Page Notice Moisture-sensitive components!
Note Duration of the data transmission: CO concentration has been set to 4. Temperature calibration procedure Mxnual calibration procedure Figure. Chapter 4 Description Of The Device