Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||19 November 2004|
|PDF File Size:||12.42 Mb|
|ePub File Size:||4.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
The unit should be returned with its cable. Complete the pulse width calibration procedure. FR A7 Overcurrent fault Verify that the error shown on the control display is within specification.
FLUKE A OPERATOR’S MANUAL Pdf Download.
Refer to Figure for the proper setup connections and connect the Calibrator Mainframe to the oscilloscope. Good static aware practices should be followed when handling this and other pieces of electronic equipment. Activates the A output if it is in standby. If the address has been changed from the factory default, change this line accordingly. These specifications are valid after allowing a warm-up period of 30 minutes, or twice the time the A has been turned off.
All other units are derived Refer to Chapter 4 of the A Service Manual for the message and a description of the possible cause. Oscilloscope Input Capacitance Measurement Specifications. Enter text from picture: An overload condition is detected. Diagnostics runs the following tests: Equipment Setup for Low Frequency Flatness. Page A configuration: The warranty period is one year and begins on the date of shipment.
Performing a Calibration Check The following tests are used to verify the performance of the A Calibrator. Leveled Sinewave Function Verification: Edge Specifications Table Remove the two rear handles by removing the six Allen screws from the handles. Sequence of Diagnostics Tests.
Fluke 5500a – Sc300 Multi-product Calibrator
At 10 ns, the user can chose between the square wave or the leveled sine signal. Chapter 9 Accessories Title Page The manua, softkey under Output toggles the impedance between 50 O and 1 MO. V optional for ac voltage and TC modes optional for ac voltage modes. To output a period of 1 ms at V peak-to-peak, use a frequency of Hz.
Watts uncertainty power voltage Testing the Front Panel. Adjusting the Leveled Sine Wave Function There are two adjustment procedures that need to be made for the leveled sine wave function. The uA range is only available in ac.
Order additional copies of the manuals or reference guides separately using the part number provided. The three types of report are as follows: Lift out the Filter PCA.
Equipment Required for Calibration and Verification Table lists the equipment, recommended models, and minimum specifications 5500q for each calibration and verification procedure. Thermocouple Measurement Accuracy Test. Equipment Setup This procedure uses the spectrum analyzer. Allow the A reading to stabilize, then enter the reading into Column A of the table. Page The memory holds calibration constants and setup parameters. In many different ways it is easy to introduce a subtle long-term stability problem by handling the boards.
Allow the DSO width reading to stabilize.
Note the RS Host port parameters that you selected in this procedure. DC Voltage Offset Accuracy. Command Syntax The following syntax rules apply to all the remote commands. Adjusting the Ledge with R16 Note Aberration adjustments are interactive with rise time adjustments. Complete List fluks Error Messages. Remove the bottom cover.
Fluke Calculator A/COIL User Guide |
The Calibrator Mainframe will prompt that the calibration is complete. High Frequency Flatness Verification at 3. Zero and self-calibrate the power meter with the power sensor being used. Cleaning the Air Filter Warning To avoid maunal of injury, never operate or power the A calibrator without the fan filter in place.
To print the manual completely, please, download it.