The Elutrap system is designed to isolate nucleic acids and proteins from agarose or polyacrylamide gel slices by electroelution. Samples are purified with . azim58 – Official elutrap documentation. Elutrap protocol. “C:\kurt\storage\CIM Research Folder\DR\\\ELUTRAP\ “. The Elutrap, seen from above, is placed on a horizontal agarose gel electrophoresis chamber. Outer limits of the trap are defined with disposable membranes at.
|Published (Last):||22 January 2017|
|PDF File Size:||7.3 Mb|
|ePub File Size:||4.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
Switch the electrodes so that the negatively charged black electrode faces the collection chamber and run for 20 seconds. Tap Tap to Zoom. Loosen the screws of the elutrap device. Take care not to damage the membranes during collection. Run the unit at V for three hours. For more brand name Laboratory Apparatus please visit our Laboratory Apparatus store section.
Info for a Biotech Student: Elutrap electro-elution system
No reviews have been left for this product. Assembly of the Elutrap system is easy. Plug in the electrodes. After collection of each sample, check the OD optical density of the buffer collected. Remember that the red colored positively charged electrode is on the same end as the collection chamber.
Place the dialysis membrane between the screw and the closed U insert in the large chamber of the device. Labnet E 35 well comb for use with 15cm gel system, 1. Gel pieces containing the purified template. The Elutrap system is designed to isolate nucleic acids and proteins from agarose or manyal gel slices by electroelution.
Elutrap Electrophoresis Chamber
Align the tray such that the two apertures lie on top of each other allowing the passage of current into the channel. Sunday, September 30, Elutrap electro-elution system.
The buffer in the electrophoresis chamber should cover the electrodes. Careful quick collection is desirable. Ultrapure TBE buffer — about 15 ml per device used. Please change the country on your eljtrap in order to switch to another country store.
BT1 Membranes Whatman No.: Collect using a loading tip, moving the tip carefully along the edges to kanual the entire sample. Leakage can be detected by the appearance of water on the paper filter.
Whatman Elutrap Electroelution System, Whatman 10404090 Replacement Components BT1 Membranes
elutrxp Wash well with warm water, followed by distilled water and DI water. We offer only genuine Whatman products, authorized for sale in the U.
Simple to Use Assembly of the Elutrap system is easy. If you cannot enable cookies in your browser, please contact us — we are always here to help! It is essential to then autoclave the buffer to remove all traces of DEPC. Keep the dialysis membrane moist till the placement of the BT1 membrane. The Elutrap system can be used with other horizontal gel electrophoresis chambers.
Allows up to four different samples to be run simultaneously. View my complete profile. Sample can be extracted from the buffer by ethanol precipitation and lyophilization.
Tighten the screws lightly. Elutrap Electro-Elution System Introduction Elutrap electro-elution system is a membrane trap elution system used for the extraction, concentration and dialysis of Eluttap and other charged molecules above the size of 5 kDa 1. Your privacy is important to us, and any personal information you supply to us is kept strictly confidential.
BT2 membrane — one per each device used. Our site is powered by industry leading security standards for your protection. The membrane placement is adjustable, eoutrap final elution trap volumes to be optimized for the particular assay. Web browser based cookies allow us to customize our site for you, save items in your cart, and provide you with a great experience when shopping LabPlanet. Please add items to your Cart for the correct pricing.
We are your premier online distributor for Whatman Laboratory Elurap. To see our complete line of lab, medical, industrial, and scientific products, please see our science and lab equipment online catalog.
Elutrap Electrophoresis Chamber – GE Healthcare Life Sciences
Please Enable Your Browser’s Cookies Functionality Cookies are not currently enabled in your browser, and due to this the functionality of our site will be severely restricted. Do not crowd the gel pieces on either end of the elution chamber. It is very important to place the sloping surface of the BT1 membrane next to the triangular mark on the side of the device.
Conclusion Organic solvents may damage the device and should be removed promptly. The OD of subsequent collections should be lower as the concentration of the sample gets diluted with the buffer. I have a saved copy of the old protocol, and if anyone needs it, feel free to get in touch with me.
There’s no available product features to display for this product variant. To check for the proper placement of the two membranes, place the device upright with the dialysis membrane on top, add some DI water and check for signs of leakage.
Prices shown may not reflect the correct price. The same applies to handling of the equipment.