The Chroma Milliohm Meter is Chroma’s newest digital Milliohm Meter. With a basic accuracy of % the instrument offers a. mΩ ~MΩ wide. On Sale Chroma ✅ Engineers on staff. ✅ FAST quotes ✅ We accept POs. A Four Terminal Test Cable. The Chroma Milliohm Meter is. Chroma’s newest digital Milliohm Meter. mΩ~MΩ wide measurement .
|Published (Last):||27 January 2007|
|PDF File Size:||15.76 Mb|
|ePub File Size:||5.75 Mb|
|Price:||Free* [*Free Regsitration Required]|
PT Temperature Probe A Unless rejected in some way, the thermoelectric EMFs can interfere with the measurement. The number of samples to average can be set from 01 to Using the A Temperature Compensation Card with A PT Temperature Probe, res i s tance values measured at ambient temperature can be corrected by applying a thermal coefficient so that chrom display shows the corresponding resistance values at any other temperature with temperature correction function.
Generally used for automatic equipment timing adjustment or delay time for waiting a real leakage current. All trademarks that appear on this website are the property of their respective owners.
CHROMA | mOHM METER |
DC, Pulsed, 116502 Dry-circuit test current driving modes, enable the Chroma can be properly used in DC resistance measurement for various inductive components coil, choke, and transformer winding etc. Conductivity Evaluation in Product Design? R Scan function should be coordinated with purchased option A 4 channels R scanner.
Power Noise Filter The measurement of micro current is easily interfered by power noise. Thus, test if the coils balance is needed.
R Scan 4 channels R scan with balance check function for fan motor combined with A option. Setting For example, if the chromz temperature is 30?
The measurement of micro current is easily interfered by power noise. For some large inductance component testing, trigger delay time must be set to delay the measurement start point till test current is ready.
DC test current output mode is used to fasten measurement speed for inductive DUT. The limit can be set to single Upper or Lower, and both, depends on the 165502 requirement.
Generally used for DUT need more time for charge. DC,Pulsed,and Dry-circuit test current driving modes,enable the Chroma can be properly used in DC resistance measurement 61502 various inductive components coil, choke, and transformer winding etc. Key Features Basic accuracy: This function avoid users complex calculation and calculation error. Average Measurement values can be averaged to minimize display instability.
Measurement values can be averaged to minimize display instability.
Chroma 16502 Milliohm Meter
The device then calculates and displays the increase in temperature. This function is especially useful for verifying motor windings or coils, where the maximum temperature increase needs to be determined when current is applied. Temperature Compensation Card A Delay time between the test start and the external trigger. The Chroma provide temperature correction function.
It sets the thermal coef? Temperature correction Temperature correction function without regard to material or temperature. Allow the motor or coil to reach room temperature, and then measure the resistance r0 and ambient temperature t0 before supplying power to the motor or coil.
Used Chroma for sale by Axiom Test Equipment |
The Chroma limits this open-circuit voltage level to be lower than 20mV. It is also possible to use the calculated value for making comparisons.
Users usually get different resistance value with different ambient temperature. The temperature conversion function can not be used at the same time as the temperature 1602 function.