The following alloys meet the ASTM-B standard. Select an alloy for product specifications: C Copper Alloy C is the most commonly specified. ASTM B Standard Specification for Copper-Beryllium Alloy Plate, Sheet, Strip, and Rolled Bar. Material Highlights: Material Highlight: ASTM-B Alloy C This material is a copper base alloy with both high strength and high conductivity attributes.
|Published (Last):||8 October 2005|
|PDF File Size:||20.10 Mb|
|ePub File Size:||6.4 Mb|
|Price:||Free* [*Free Regsitration Required]|
Add two drops of HCl after the? Allow the crucible to cool, add asttm mL of water, and? A superscript epsilon e indicates an editorial change since the last revision or reapproval. Cool, dilute to volume, and mix. This standard has been approved for use by agencies of the Department of Defense.
Asym the instrument be equipped for multi-point calibration, make sure that several additional solutions still are analyzed to ensure that error has not been introduced by the curve-? The absorption of the resonance line energy speci? An acidic solution of silicic or?
Mechanical Property Requirements 8. Ignite the burner and aspirate water until the instrument reaches thermal equilibrium. Exercise care to prevent contact of beryllium-containing solutions with the skin. The following alloys are speci?
Cover the beaker and cautiously add mL of nitric acid HNO3 in small increments. The determinations are made on the separate samples and in a plane perpendicular to the surface and perpendicular to the direction of rolling. Precipitation heat-treatment parameters for these bend tests shall be zstm accordance with This solution shall be freshly prepared as needed. Therefore, it is necessary to have the same amount of copper, 6 mg, present in the?
ASTM B194 Grade C17000 TM01
For the purpose of determining conformance to speci? Cautiously add 35 mL of HNO3 in small increments. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard. Allow to stand until dissolution is nearly complete.
Precipitation heat-treatment parameters for Table 2 and Table 3 are shown in Section If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below.
E 29 Practice for Using Signi? E The thickness of material that may be tested by use of the Rockwell Hardness scales is as follows: In all cases only the current version of the standard as published by ASTM is to be considered the official document.
Add 20 mL of the acid mixture. Specimens shall adhere to specified chemical composition, grain size, and grain count requirements. This specification establishes the requirements for plates, sheets, strips, and rolled bars of copper-beryllium alloy with Copper Alloy UNS Nos.
Repeat all measurements at least two times.
It is expected that work will be performed in a properly equipped laboratory. A uniform for color development should be established and then used for both calibration and test solutions.
When dissolution is complete, add 25 mL of boric acid solution to three mL plastic? Store in plastic container.
ASTM-B Alloys | E. Jordan Brookes Company
Hardness values shown apply only to direct determinations, not converted values. Cool, wash down the cover glass and inside of the beaker. Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters.