This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||1 September 2009|
|PDF File Size:||1.83 Mb|
|ePub File Size:||1.75 Mb|
|Price:||Free* [*Free Regsitration Required]|
Please read it and become familiar with its terms. File Menu Functions File Menu Functions This chapter describes how to use the functions located under agllent front-panel Data storage and retrieval are handled similarly to that of personal computers PCs.
Agilent Technologies E4440A Getting Started Manual
Options And Accessories Options and Accessories This chapter lists options and accessories available for your analyzer. This manual also for: Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: In this example you are renaming a trace file; press 3.
Entry fields that are being modified or are modifiable are indicated by highlighting inverse video. If you wish to verify this condition, remove the signal input. Installation And Setup Installation and Setup This symbol indicates the input power required is AC. To see that trace 2 is in view mode, blank traces 1 and 3 by pressing.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
AC Power Cords Chapter Page 25 Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation In Case Of Difficulty In Case of Difficulty This chapter includes information on how to check for a problem with your Agilent Technologies spectrum analyzer, and how ahilent return it for service.
File Catalog Figure Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
Table of Contents Add to my manuals Add. To access the Load menu, press 3. To print the manual completely, please, download it. Restores the analyzer to its factory-defined state.
Page 80 Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters The analyzer was shipped from the factory with the Alignments mode set to Auto.
Page 23 Item Name Access e4440a used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Additional Information Notice The information contained in this document is subject to change without notice. A pound uncoupled function.
f4440a The file cannot be placed inside another file. Protecting Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used. Refer to Figure Activate trace 2 and put it into minimum hold by pressing analyzer display should look similar to Agioent Power Requirements Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source.
Press If the previously performed alignments did not resolve the problem, press Alignments Restore Align Defaults Is the analyzer displaying an error message? Contacting Agilent Technologies Online assistance: Put a formatted floppy in the A: Page 44 Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: Viewing Traces 1, 2, and 3 Step 2.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
User Preset This type of preset restores the currently selected mode to a known Mode Preset The entry fields show the parameters for the files viewed. Viewing A Signal Viewing a Signal 1.
Agilent Technologies makes no warranty of any kind with regard to this material, including but not limited to, the implied warranties of merchantability and fitness for a particular purpose.
If the softkeys Preset Factory Preset 2. Copying a File 1.
The battery is maunal on the CPU board. In this example you will save a trace to the internal C: Knob Increments or decrements the current value.
In Case of Difficulty Returning an Analyzer for Service Returning an Analyzer for Service Service Tag When you are returning an analyzer to Agilent Technologies for service, fill out and attach one of the blue service tags provided at the end of this chapter. For a complete list of spectrum analyzer messages, see the Instrument Messages and Functional Tests manual.
This e4440aa can be auto-coupled.
Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components msnual transported, stored, or used. The analyzer has three types of preset: Like PCs, these analyzers include an internal storage drive and a floppy e4440s drive, both of which have directory and sub-directory capability.
Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation