to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||25 July 2009|
|PDF File Size:||2.11 Mb|
|ePub File Size:||3.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Cleaning Cleaning Cleaning A clean optical measuring system is a basic prerequisite for determining precise measured values. By pressing ‘print’ button you will print only current page. Error messages Error messages In certain circumstances error messages may appear on your display.
Safety Information Safety information Protection against infection There is a potential risk of infection. Switching On The Instrument Performing a measurement Switching on the instrument Place the instrument on a level surface or hold it in your hand.
Quality control recommendation Quality control recommendation To ensure that the instrument is functioning properly, you msnual regularly use two levels of control solutions for a function check.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
If you are measuring glucose, perform the plausibility check described on the next page. Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip.
Internal error example shown: Page Alphabetical index Operating the instrument The flap firmly locks into place when it reaches a vertical position. Are date and time correct? About this manual Example of a display screen: Sample Material Performing a measurement Sample material Fresh capillary blood is used as sample material.
Performing A Control Test Quality control recommendation Performing puls control test Place the instrument on a level surface or hold it in your hand. Page 99 Cleaning If needed, you can rinse the strip accutrnd separately from the instrument under warm running water. Page 53 Performing a measurement Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements.
Quality control recommendation After switching on manuwl instrument, check the following displays: Temperature The instrument or ambient temperature does not fall within the acceptable range for the selected test. Page 62 Performing a measurement Now take the test strip from the test strip container. Starting The Measurement Performing a measurement Starting the measurement 13 Close the measurement chamber flap.
Make sure that your assignments are unique and reproducible.
Test Principle Introduction Test principle By means of a code strip, the instrument reads the lot-specific characteristics of the test strips currently in use. Press the M button until the desired month is displayed. Performing a measurement The three-digit code number which is also printed on the reverse side of the code strip and on the test strip container is shown in the display.
Got it, continue to print. If the problem persists, please contact your local customer support and service center.
Flagging Measurements Performing a measurement Flagging measurements You can add additional information to measurements, for example, to characterize a particular value with regard to special circumstances e.
Corrections can only be made by repeating the settings. Alphabetical Index Alphabetical index Alphabetical index Capillary blood With a set of fresh batteries, you will normally be able to perform approximately 1, measurements.
Cobas Accutrend Plus User Manual
Performing a measurement A flashing arrow now instructs you to open the measurement chamber flap to apply the blood. When the instrument turns itself off, all test results obtained so far remain in the memory. Therefore, clean the instrument regularly and immediately after it becomes dirty.
Page 30 Operating the instrument Performing a measurement Blood application in the instrument: Check that the pllus functions correctly by comparing it to the diagram on the left to prevent misinterpretations due to defective display elements. Further information Product specifications Operating conditions and technical data Temperature range for Cholesterol: Memory Memory Memory The Accutrend Plus instrument has two memory areas, each of which can be used to store up to measured values together with date, time and event flags.
Don’t show me this message again. To run a test, plue unused test strip is taken from the container and inserted into the instrument. Page 87 Quality control recommendation Open the measurement chamber flap.